Is the cleaning procedure any different for this new injector?
The procedure is the same. However, the WB Injector needs more frequent cleaning than the HT Injector. It is recommended that after each day (8 hours) of acquisition, the injector be rinsed with distilled, deionized water and dried thoroughly before you perform additional experiments. Weekly cleaning with 10% Decon™ Contrad® should still be performed following the instructions in the Helios™ User Guide (PN 400250).
The Sampler and Skimmer-Reducer Cones need more frequent cleaning as well, after approximately 12 hours of running with Maxpar® Cell Acquisition Solution as carrier. Follow the standard procedure for cone cleaning as described in the Helios User Guide.
For Research Use Only. Not for use in diagnostic procedures.