Why is there a 10 Ct shift from tube-based qPCR to the Biomark system?
The difference in volume between a 10 μL reaction and the 96.96 Dynamic Array IFC with a 6 nL chamber is 210.7. For the 48.48 Dynamic Array IFC this difference is 210. This predicts that a single copy will occur at approximately 10.7 to 10 cycles lower for the IFCs than when using standard microcentrifuge tubes with 10 μL reactions.
For Research Use Only. Not for use in diagnostic procedures.